Throw-in Type Liquid Level Transmitter FAQ【KIT0139】
Q1:Why are there some differences in readings at different temperatures?
A1:The sensor uses silicon components, which will have a certain temperature drift phenomenon.
For more accurate measurement, it is recommended to use a temperature close to 25℃.
Q2:Why is there a discrepancy between my reading and the actual depth?
A2:The wiki maps the measurements for user understanding, but the sensor probe's sensing diaphragm is not at the bottom, but near the nut.
The error range of the sensor is 5000*0.5%=25mm, which is suitable for some scenarios with larger depth.
Q3:The sensor intermittently outputs very high values during measurement.
A3:Please check whether your power supply is stable, this phenomenon is due to the clutter value generated by the power supply.
If you have other questions, just feel free to post them in our forum for consultation. We will reply them as soon as possible and improve the product FAQ gradually.